PIECAL 820 Elite
Carry a whole toolbox of single function calibrators in the palm of your hand with the PIECAL 820 Elite, from Practical Instrument Electronics (PIE).
Pressure (via optional external pressure module)
The intuitive EZ-DIAL Double Click Menu makes it easier to setup than many other multifunction calibrators and uses the same menus as the single function PIECAL Evolution Calibrators. Icons on the display indicate where to plug in the test leads.
Power up transmitters and loops
Simulate 2 Wire transmitters
Read to 24.00 mA, 0.0 to 60.0 VDC, +/- 99.999 mV and +/- 999.99 mV
Source 0.00 to 24.00 mA, 0.00 to 10.25 VDC, -20.000 to 99.999 mV and -500.00 to 999.99 mV
Use it as a milliamp and voltage calibratorwith
built inSimplify HART
24 V power supplyhook ups with
built in 250 ohm resistor
Calibrate directly in temperature (°C & °F)
The PIE 820Elite works with the instruments you use.
Thermocouple types J, K, T, E, R, S, B, N, G, C, D, L (J DIN), U (T DIN) and P (Platinel II)
RTD types Pt 100 Ohm (3850, 3902, 3916, 3926) & 1000 Ohm (3850)
Copper 10 & 50 Ohm, Nickel 110 and 120 Ohm
Checkout flow and vibration systems
Source & read frequency to 2000 CPM (Counts-Per-Minute), 999.99
Hz, 9999.9 Hz, & to 20.000 kHz
Troubleshoot loop & wiring problems
Beep out connections with the built-in continuity checker
Simulate pH probes into transmitters & analyzers
Source from 0.000 to 14.000 pH @ 25°C (77°F) corresponding to +414.12 to -414.12 mV
Measure pressure with optional pressure modules
32 modules for gauge, differential, compound and absolute pressure
Easy to read
Turn on the backlight & easily see the display in dark areas of the plant
Quickly set any three outputs plus automatic stepping & ramping
Easily set any value quickly with the adjustable “DIAL”; store any three output settings for instant recall with the EZ-CHECK™ switch.
Choose between 2, 3, 5 & 11 steps to automatically increment the output in 100%, 50%, 25% or 10% of span.
Select RAMP to smoothly increase and decrease the output between Zero and Span.
Set step/ramp time to match your system from 5, 6, 7, 8, 10, 15, 20, 25, 30 and 60 seconds
Measure temperature sensors, frequency pickups, loop currents, voltage levels & pressures
Check the values of your process sensors. Instantly recall MAX and MIN values to see process variability.